Turn on the front-panel power switch. Consult your network administrator if you do not know whether your network uses DHCP. With this test panel, you can set properties for your instrument communication as well as read and write to the instrument. To correct the problem with the hostname, complete the following steps:. The unit is now ready to operate. The current hostname is displayed.
|Date Added:||10 March 2011|
|File Size:||42.21 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
This indicates that you should check enett loose or disconnected cables, power disruption, or a malfunctioning instrument. Unfortunately, engineers sometimes forget to reconnect all the cables and power up all the instruments. You may need to run software configuration and verification utilities at this time.
National Instruments GPIB-ENET/ – In Stock, We Buy Sell Repair, Price Quote
Also have the network administrator assign the proper subnet mask, gateway, and DNS server addresses. Then select your GPIB controller. For 10000 subnet mask, numbers between 1 and are valid choices for the last number of the IP address. This menu displays the list of supported hardware for the installed version of NI It makes GPIB instrument detection and control easy by providing tools that help you search for connecting instruments, and send and receive communication with your device.
Connect the other end of the Ethernet cable to your Ethernet network. If your network uses DHCP, the network configuration is performed automatically at startup, and you do not need to run this utility unless you want to change the hostname.
These productivity tools help you focus on developing your test applications instead of trying to track down obscure problems or establishing communications with your instruments.
Once you have determined the GPIB address of your instruments, it is easy to establish communications to dnet that you can send and receive data to and from the instrument.
ID querying is discussed in the next section.
If it is not, then you know to check cables, confirm power, and verify that the instrument is working properly. Using your computer, you can take advantage of this powerful development and debugging tool to interactively communicate read, write, serial poll, and so on with your GPIB instruments. The unit is now ready to operate.
Installing GPIB-ENET/100 Controllers
Instruments must be powered on and connected to the GPIB controller to be recognized. Click on Scan for Instruments. Exit when you are finished using the Ethernet Device Configuration utility. You also can use the Resource to Find field to query each instrument for an identification ID string. The basic functions and concepts described above may seem simple; however, they can be invaluable in troubleshooting and getting your GPIB system up and running.
The utility alerts you if it detects a problem with the network settings. Connect the other end to your GPIB instrument. The Ethernet address is not the IP address. gipb
Run the Ethernet Device Configuration Utility. In addition to manually configuring the network parameters, you can use the Ethernet Device Configuration utility for any of the following gpbi.
National Instruments GPIB-ENET/1000 GPIB Instrument Control Device
You can repeat this process to confirm communication with all of your connected instruments. If your network uses DHCP, the network configuration is performed automatically at startup.
View the properties for the device. The format of the IP addresses is determined by the subnet mask. It is important to choose an alias that you can immediately identify with the intended instrument.
If you have already seen and closed the initial NI Refresh the list of devices, after the device reboots, to verify that the hostname is now valid.